72 Posts in 31 Topics by 15 membersJump to:
Welcome to Decagon's online forum. Find answers to common questions for Decagon personnel and other users. If this is your first visit, you will need to register before you can post. However, you can browse all messages below.
|Page: 1||Go to End|
|Author||Topic:LWS sensor conversions||1998 Views|
16 October 2010 at 4:17am
I have been using the Leaf wetness sensor, the manual states how to convert raw counts to mV can this be then convorted to mL?
18 October 2010 at 4:20pm
We have done some exploratory research to see if the leaf wetness sensor can be used quantitatively to determine the amount of water on the leaf. You can use this equation:
mg H2O/cm^2 = 0.047*rawADC - 20.9
This equation fits the data with an r^2 of 0.97. It was done in the laboratory and has not been tested in the field or over different temperatures to our knowledge.
23 October 2010 at 6:05am
Thank you very much for the information,
It has veen very useful to be able to convert the data into ml.
|Go to Top|
Currently Online: There is nobody online.
Welcome to our latest member: lafseo